Chamber furnace СНО-2,5.5.2,5/11 И1
Electric furnace CHO-2,5.5.2,5/11 И1 is made on special customer requirements for specific products and heat treatment has a non-standard solution in the door is opened. Since 1994, OOO "Bortek" produces a variety of electric furnace for a constructive solution and for their intended purpose.
|I1 - outdoor heaters inside the oven chamber|
|1||Rated power, kW||
|Power supply||3×380 V, 50 Hz|
|Maximum temperature in the working space, ºС||
|4||Range of automatic temperature control, ºС||40...1100|
|5||Dimensions of working space, length × width × height, mm||250 × 500 × 250|
|6||Atmosphere in the working space||Air|
|7||The stability of the temperature control in the steady thermal conditions, without cages, ºС||± 3|
|8||Type of temperature regulator / thermocouple type||TRP-09-TP / THA|
Chamber laboratory electric furnace CHO-2,5.5.2,5/11 И1 is equipped with an automatic temperature control using contactless relay and solid-state digital temperature control, allows you to specify complex modes of heat treatment ponds - 12 portions rise-shelf and has access to a computer with the interface and software. The control panel is placed on the left side of the furnace. Open-frame thermocouple is located on the rear wall of the furnace. Chamber furnace is made of microporous and fibrous refractory and thermal insulating materials having low values of the heat capacity and thermal conductivity. On the upper side of the furnace near an end wall and the bottom of the door has a vent opening slug to remove any gaseous discharge.
On the doors and on the rear wall of the furnace made three holes with a diameter of 20 mm with removable plugs. The centers of the holes are located at a height of 125 mm from the furnace bottom. In the center there is one hole relative to the side walls of the furnace, and other openings are located at distances of 35 - 40 mm from the center hole. Framed hole ceramic tube and the outside of the metal sleeves. Doors move on sleeves on two guide rods along the longitudinal axis of the furnace at a distance of 500 mm. The guide rods are attached on one side to the front side of the oven door and have, at the opposite end, bearing on the bearing surface on which the furnace.
Heaters of fehrali placed on the tubes in the recesses of the side walls of the furnace and beneath the floor of the furnace. All heaters are open. The furnace power is increased to ensure rapid heating furnace. Keep foreign objects, including scaling on heaters.
Write a reviewYour Name:
Your Review: Note: HTML is not translated!
Rating: Bad Good
Enter the code in the box below: